We have a new knowledgebase at kb.egauge.net!
This article can be found HERE.
The eGauge has the ability to back up data recorded on the device, and restore it to any eGauge device with the same register configuration. This includes eGauges from different hardware revisions - for example, a backup file from EG30xx hardware can be applied to EG4xxx or eGauge2 hardware. The register configuration must be identical (including spelling, capitalization, spacing, etc). Registers that do not exist on the destination device will be ignored, and registers that do not have a match in the backup file will be overwritten with zeros.
For saving data to a spreadsheet, please see our video tutorial on exporting data as a CSV file.
The backup function is found in Tools > Backup. From here, you can choose whether to perform a full backup, or partial backup. If performing a partial backup, choose the time range for the data to be backed up from. The backup file will be named backup-DEVNAME.bin where DEVNAME is your eGauge device name. This file is only used for restoring data to the eGauge; it cannot be processed or modified. CSV and .bin files are not interchangeable.
To restore a backup file, navigate to Tools > Restore. From here, you can choose a full restore (all data in the backup file is restored), or a specific range. When choosing full restore that the backup file will overwrite any data on the device. This means if the backup file covers a date range where the eGauge has no data, this blank data will overwrite potentially valid data in the same time period on the device.
If uncertain about how much data is in the backup file, only restore the range of data necessary (ie, the data that is missing).
If restoring fails or hangs, please attempt the restore from the same LAN the eGauge resides on. This is done by using a computer connected to the same network as the eGauge, and clicking on LAN Access in the upper right-hand link menu before performing the restore.